Introductions of FD-4 Electric Gynecological Operating Table:
Electric Table gynecologic surgical for maternity and use a variety of gynecological surgical medical apparatus and instruments, before and after the surgical table mesa, left and right sides, lifting up and down and back to fold up and down all by electric putter, button operation, safe and reliable. The table by the main table and table leg table in two parts, the table body by the oil pump of hydraulic adjustment, leg panels are removable as well as outreach, convenient adjustment, is made from double deck plate with high strength through plate, can be ray diagnosis at the time of surgical
|Adjuestable range of height||1000~750mm|
|Back section raised from the horizontal||≥ 75 °|
1，Click on the power grab switch on the controller panel to enter the preparation stage.
2，Correctly start and release the base brake, the brake pads next to the lower base, and move to the fixed bar. In order to set up the operating table with the central mechanical brake.
3，The hand should be hung on the side rail of the operating table. Its circuit should avoid face, press, and prevent the circuit from bad.
4，Do not place heavy objects on the power cord or push the cart over the power cord.
5，Do not let the patient sit on the head plate of the operating table, the hand valence plate or the leg plate, heavy weight can cause the fittings to bend the report bad. The headboard and the longboard are the maximum of 40kg. When two legs apart more than 45 ° can load only 20 kg. The operating table should bear no more than 150kg.
6，Do not place items, accessories or weights on the outer cover of the base of the operating table.
7，The cleaning and disinfection of the operating table and accessories. Use the weak glass with surface active tree and phosphoric salt to clean up the operating table and accessories. The operating table disinfection should be used surface disinfectant containing b, tsuen wan, limiting dilution table, can't use, scratching, containing compounds and mixtures containing b drunk, so as to avoid corrosion of metal surfaces. Do not use cleaner and water to spray or rinse the base to prevent the electrical control system from being short-circuited and the parts rust or fail.
8，Do not operate oil pressure motor for more than 5 minutes continuously to avoid failure.